Original filename: 5002.1000X.pdf
Title: Microsoft Word - 5002.1000X
This PDF 1.4 document has been generated by PScript5.dll Version 5.2.2 / Acrobat Distiller 10.1.0 (Windows), and has been sent on pdf-archive.com on 21/08/2017 at 09:12, from IP address 72.52.x.x.
The current document download page has been viewed 724 times.
File size: 175 KB (1 page).
Privacy: public file
Download original PDF file
5002.1000X.pdf (PDF, 175 KB)
Share on social networks
Link to this file download page
Page 1 / 1
Measuring range: 500 -1100 hPa
Resolution: 0.01 hPa
Accuracy: ±0.2 hPa at ambient temperature
Long-term stability: ± 0.2 hPa / year
Operating temperature: -40...+60°C
Supply: 7...30 V DC, < 3 mA
Output: 0...5 V DC
This product complies with European CE requirements for the EMC Directive.
Dimensions: approx. 100 x 60 x 22 mm
Weight: approx. 140 g
5002.1000X is a high-precision Barometric Pressure
Sensor offers unparalleled performance and flexibility
and combines high accuracy and low power over a
wide range of pressures and temperatures for atmospheric pressure measurement.
This Barometric Pressure Sensor provides a calibrated 0-5 VDC analogue output and its analogue current
consumption is less than 3 mA. A special “sleep”
mode further reduces current to about 1 μA.
The compact size of the barometer allows easy
placement in most standard instrument enclosures. A
plug-in connector simplifies wiring.
The 5002 series barometers are designed to be
installed indoors or inside equipment only. The
barometers can also be mounted on a 35 mm
wide standard DIN mounting rail by using a
mounting foot and screw supplied with barometers.
The barometers should be installed vertically with
the connectors downwards to prevent any ingress of condensated water. Horizontal installation can be used under conditions where no condensation can take place.
Note: 1 hPa = 1 mbar
Hectopascal (hPa) is the barometric pressure unit
recommended by WMO and also accepted by ISO.
See WMO Guide to Meteorological Instruments
(1983) and ISO 1000:1992 (E) for further details.
Technical data are subject to change!
THEODOR FRIEDRICHS & CO.
Meteorologische Geräte und Systeme GmbH
POB 1105 D 22858 Schenefeld Tel. (040) 839 600-0 Fax (040) 839 600-18
Email: firstname.lastname@example.org http://www.th-friedrichs.de
Link to this page
Use the permanent link to the download page to share your document on Facebook, Twitter, LinkedIn, or directly with a contact by e-Mail, Messenger, Whatsapp, Line..
Use the short link to share your document on Twitter or by text message (SMS)
Copy the following HTML code to share your document on a Website or Blog